So we have our catheter (applied part) connected to our system (accessible part), all good so far.
Our catheter is an over-the-wire catheter, It is not part of the ME, but we use it to achieve our intended use.
Is the following correct? :
From my perspective, the guidewire will be considered a patient connection and not an applied part. Therefore I will still be required to ensure insulation between the guidewire and the system.
Indeed, your understanding is accurate. The guidewire utilized alongside the over-the-wire catheter is classified as a patient connection rather than an applied part.
Given its classification as a patient connection, it is essential to implement suitable insulation and electrical safety measures to ensure the device operates safely and effectively. It is necessary to have insulation in place between the guidewire and the system to prevent any potential electrical hazards or negative impacts during device usage.
So I can make that isolation of components. Still, at the end of the day, for defibrillation-proof, for example, if the proximal end of the guidewire (the part that is outside of the patient) is on the operating table, there is so much I can do about the measures I can put in place so that the defibrillation energy will not be delivered to the patient.
I can do that for my system, so it will be considered safe if the users accidentally touch it. But not for the entire setting, as there will always be an exposed part of the guidewire.
Regulatory Affairs Professionals Society (RAPS)5635 Fishers Lane, Suite 400Rockville, Maryland 20852
email@example.com+1 301 770 2920
JoinMy RAPS DashboardLearn More